English | PDF | 559.8 KBDownload now
Gamma radiation emitted from a shielded source travels through the vessel at the set height to a detector. When product level enters the beam path, the radiation is significantly weakened or totally bloked so that the amount of gamma at the detector suddenly drops. This change in counts per second at the detector results in the detector switching a relay or a sudden change in the 4-20 mA output. When used as a low level switch, the detector normally sees no counts from the source, and triggers the relay or changes the 4-20 mA output when level drops below the source and it suddenly sees many counts from the source.
Temperature, pressure, dust, color, and chemical properties of the product have no influence on the outcome of the measurement. Consequently this results in the high reliability and low maintenance of the radiometric measurement systems, even under harsh operating and environmental conditions.
Our highly sensitive detectors achieve a better measuring effect and operate with remarkably low source activities. It is able to measure, where other detectors fail or need to work with unacceptable high source activities.
Berthold‘s patented method for detector stabilisation ensures consistent measurement performance, reliably compensating for temperature and aging effects. If the detector drift were to go unnoticed and uncompensated, it could cause considerable loss in production yield and product quality, or necessitate frequent recalibrations.
Cosmic radiation control makes it possible to keep the required source activity very low. Only Berthold can offer a SIL2/3 detector certified for either high or low level switching, the LB 480.
The LB 480 SpeedStar version detector is used when very fast process changes must be monitored. The SpeedStar feature ensures extremely fast response times in the millisecond range. This feature is used when monitoring steel plate edge in order to straighten the cut edges while the plate is being transported on a roller conveyor, or to position glowing and scorching steel blocks. It is also used to detect over or under-filled liquid containers, e.g. gas tanks, while they are being conveyed.
With this function, weld tests will no longer cause false low level signals. The counts per second of gamma received at the detector is monitored, and when there is a sudden jump above the counts seen from the source, the measured value is frozen until the external radiation influence is over. A digital output signals the detection of interfering radiation. All our level measurements are equipped with XIP.
A whitepaper explaining how radiometric measurements work and the components, as well as how they can be optimized for the highest accuracy and reproducibility. The paper explains best practices and how proper design minimizes error.
Do you need reliable control of your
English | PDF | 1.8 MBDownload now
English | PDF | 1.8 MBDownload now
Contact our worldwide network of engineers and service technicians for expert advice and support from initial inquiry through installation and eventual source reclamation.
Use of an outdated browser
In order to make your Internet experience safer and as agreeable as possible, you should update your browser here to a new version. The update is free. If you are using this PC at work, you should contact your IT administrator. Please note that due to the outdated browser this website may not display correctly.
Click here to ignore this message.
Berthold Technologies GmbH & Co.KG
Calmbacher Str. 22
75323 Bad Wildbad
Tel.: (+49) 7081 177 0
Fax: (+49) 7081 177 100