Brochure of the Lumat LB 9510 Tube Luminometer
EN | PDF | 445.4 KBDownload now
The new Lumat is athat uses state-of-the-art technology based on more than 40 years of experience engineering and designing luminometers. The luminometer supports all basic luminescence technologies including:
Thanks to a stringent selection of photomultipliers working in photon counting mode and more than 40 years of experience building, the new Lumat offers excellent luminescence sensitivity: <1 zmol firefly luciferase and <1 amol ATP/tube (high sensitivity model) can be reliably detected.
The Lumat can be equipped with up to 2 JET injectors with variable volumes. The ultra-fast injection speed of JET-injectors eliminates any delay between injection and measurement and is critical to measure fast reactions, e.g. for flash luminescence measurements.
Berthold Technologies' JET injectors use Teflon bellows for accurate and fast injections and guarantee most efficient mixing as well as extreme longevity.
The measurement of individual samples one after the other can be time-consuming. The Lumat has a turret mechanism that makes this job easier and increases your throughput. Thanks to its motor-driven revolving magazine mechanism based on a rotating chamber for two tubes (one for measuring and one for loading and unloading), so you can prepare and insert the next sample while the previous one is still measuring.
is a flexible software enabling easy data acquisition, analysis and reporting using Berthold Technologies' instruments. It has been optimized for different microplate and tube formats. LightCompass® features all the functions you can expect from an advanced scientific software, such as flexible creation of measurement protocols, curve fitting, qualitative analysis and compliance with FDA 21 CFR Part 11.
Low-noise photomultiplier tube in single photon counting mode
|Measurement modes||Endpoint, Kinetics|
|Spectral range luminescence||380-630 nm|
Up to 2 JET injectors with frictionless operation
10-100 µL or 25-300 µL
Better than 2% across the entire volume range
Better than 2% across the entire volume range
Luminescence tubes, 12 x 47 mm
Luminescence tubes, 12 x 55 mm
Luminescence tubes, 12 x 75 mmMicrocentrifuge tubes, 1, 1.5 and 2 mL
Standard model: <5 amol ATP/tube
High-sensitivity model: <1 amol ATP/tube
≥6 orders of magnitude
|Interface||Windows software, connected to computer by USB|
|Operating system||Windows 10 or better|
|Available software packages||LightCompass®: flexible software for data acquisition, analysis and reporting|
|Power Supply||100 – 240 VAC ±10 %, 50/60 Hz|
External auto ranging mains adaptor
Storage: 0-40° C
Operation: 15-35° C
Transport: –20 °C to 50 °C
|Altitude||max. 2000 m (above sea level)|
Dimensions (W x D x H)
|240 x 280 x 220 mm|
Due to its superior sensitivity of <1 zmol firefly luciferase and <1 amol ATP/tube (high sensitivity model), the Lumat is ideally suited for
and many more
Here you have a selection of recent papers using the new Lumat and earlier Lumat models.
|2023||Chisa Shiraishi, Akinobu Matsumoto, Kazuya Ichihara, Taishi Yamamoto, Takeshi Yokoyama, Taisuke Mizoo, Atsushi Hatano, Masaki Matsumoto, Yoshikazu Tanaka, Eriko Matsuura-Suzuki, Shintaro Iwasaki, Shouji Matsushima, Hiroyuki Tsutsui, and Keiichi I. Nakayama||Nature Communications||Dual Luciferase|
Wing-Hang Ip, Britta Wilkens, Anastasia Solomatina, Judith Martin, Michael Melling, Paloma Hidalgo, Luca D. Bertzbach, Thomas Speiseder, and Thomas Dobner
Minjie Zhao, Wei Wang, Ya Lu, Nan Wang, Delei Kong and Lina Shan
Molecular Medicine Reports
Hidehiko Kikuchi, Kaori Harata, Harishkumar Madhyastha and Futoshi Kuribayashic
Biochemistry and Biophysics Reports
Measurement of O2− generation using Diogenes-luminol probes
Rahmi Imamoglu, David Balchin, Manajit Hayer-Hartl, and F. Ulrich Hartl
Weijia Luo, Yuan Dai, Zhishi Chen, Xiaojing Yue, Kelsey C. Andrade-Powell & Jiang Chang
Yajing Liu, Yinghua Hao, Yuxia Li, Yadan Zheng, Jiajing Dai, Fubo Zhong, Wei Wei, and Zhengyu Fang
Qingchen Zhu, Tao Yu, Shucheng Gan, Yan Wang, Yifei Pei, Qifan Zhao, Siyu Pei, Shumeng Hao, Jia Yuan, Jing Xu, Fajian Hou, Xuefeng Wu, Chao Peng, Ping Wu, Jun Qin, Yichuan Xiao
Journal of Experimental Medicine
Jae-Wook Lee, Hong Sug Kim, and Eun-Yi Moon
BioLux® Gluc Gaussia Luciferase assay
You can find additional publications which have used the Lumat and older Lumat models in publication databases.
Please keep in mind that the list provided is not exhaustive, but it offers valuable examples of experiments performed with our instruments and published in peer-reviewed journals.