Moisture, dry substance and concentration with microwave measurements
English | PDF | 1.8 MBDownload now
The LB 567 is designed for applications with constant measuring conditions (e.g. constant particle size, bulk density etc.). The measurement is carried out without the need for radiometric compensation. With a height sensor, the system compensates for varying bulk heights on the conveyor belt. For more demanding applications however, we recommend the use of the MicroPolar Moist LB 568.
The moisture inside bales is a critical parameter for the quality and durability of lucerne. Material which is too moist can start decomposing, with the risk of self-ignition. A moisture measurement is carried out on the bales to make sure that the moisture content is within the desired range of 8...15 %. When Lucerne is used in Bio-Ethanol production or power generation, the online measurement of moisture is an important factor for an efficient process.
The moisture measurement analyser MicroPolar Moist LB 568 is used when the product bulk density varies. In addition to the moisture measurement with microwave technology, the system also uses a radiometric source and detector to measure the area weight (mass per unit area). This combination, allows the microwave moisture signal to be compensated for changes in bulk density, resulting in significantly improved measurement accuracy.
Use of an outdated browser
In order to make your Internet experience safer and as agreeable as possible, you should update your browser here to a new version. The update is free. If you are using this PC at work, you should contact your IT administrator. Please note that due to the outdated browser this website may not display correctly.
Click here to ignore this message.
Berthold Technologies GmbH & Co.KG
Calmbacher Str. 22
75323 Bad Wildbad
Tel.: (+49) 7081 177 0
Fax: (+49) 7081 177 100